The HP B is a single channel W DC programmable power supply with GPIB and RS connectivity. It is part of the HP x series. Agilent B/B/B/B System DC Power. Supplies All references to HP-IB have been changed to GPIB. A note about. I just picked up two HP B power supplies as I read somewhere quickly that it not only can be programmed via GPIB to set voltage/current.
|Published (Last):||26 March 2016|
|PDF File Size:||10.3 Mb|
|ePub File Size:||18.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
B – Verification and CalibrationTable B- 5. Unless otherwisenoted, both units will be bp to by the description “dc source” throughout this manual. Shopbop Designer Fashion Brands. Parameter lists are circular; you can return to the startingposition by continuously pressing either key.
Page 43 Front Panel Operation – 52 – Querying and Clearing Output ProtectionThe dc source will disable its output if it detects an overvoltage or overcurrent fault condition. Enter the existing password from Entry keypad and press Enter3. Warranty information is printed in 663b2 front of this guide. Bench OperationA fan cools the dc source by drawing air in through the sides and exhausting it out the back.
How much was the shipping on the Ebay?
The fault FLT output,also referred to as the DFI discrete fault indicator signal in the front panel and SCPI commands,is an open collector circuit that pulls the positive output low with respect to the negative chassisreferenced common. Specifications – ATable A Scroll through the Output menu by pressing q. Wait for hl DMM readingto stabilize.
Ebay buyers of HP supplies take note: ProtIndicates that the output has been disabled by one of theprotection features. The display hhp to Meter mode when the calculation is complete. Otherfunction keys have commands underneath them that are accessed when the key is pressed. Minimum clearances for bench operation are 1 inch 25 mm along the sides. Amazon Renewed Refurbished products with a warranty.
HP (Hewlett-Packard) Power Supply B User Guide |
Mail will not be published required. You do 6623b have to do a complete calibration each time. Turn off the dc source and connect it as shown in figure B-1Bwith the ohm load resistor. Table B-1 lists the equipment required for calibration. Amazon Rapids Fun stories for kids on the go. Page 33 Turn-On Checkout – 4Procedure6. Anyhoo, the unit showed up all kinds of Dirt McGirt:. Please enter a question.
Providing the finest test equipment solutions since 1992
This capability is only available when the unit is set to the SCPI programminglanguage. Although my original post about this is pretty old I still use this same unit as my main bench supply. Or at least you can get close.
This thing is heavy!! Unscrew the line fuse cap from the rear panel and verify that the 6632g rating matches what isspecified on the FUSES label on the rear panel.
HP B : – Hewlett-Packard Power Supply User Manual
I just got it from a good friend of mine, who owned two of these fine instruments. Amazon Second Chance Pass 66332b on, trade it in, give it a second life. Connect the DMM in dc mode across the current shunt. Hewlett-Packard Asia Pacific Ltd. If you read through the manual, you know, that it can also behave as current sink load.
You can read what HP 6632bb about it but it basically kicks as much ass as I could really want or expect a single output PSU to kick.
663b 46 5 – Front Panel Operation5 – Programming the Output Relay option only 6632b with option have isolation and polarity reversal relays connected to the output and senseterminals. Figure B-1 shows the test setup. When it was done, the front panel meters were dead on, and the output voltage was within 0. Set the output current to its full scale value see Table B Function KeysImmediate Action KeysImmediate action keys immediately execute their corresponding function when pressed.
ComiXology Thousands of Digital Comics.